SPE Syringes | SHIMADZU EUROPA
SAMPLE PREPARATION – MEPSTM
MEPSTM is Micro Extraction by Packed Sorbent and is a development for sample preparation and handling. MEPS is the miniaturization of conventional SPE packed bed devices from milliliter bed volumes to microliter volumes. The MEPS approach to sample preparation is suitable for reversed phases, normal phases, mixed mode or ion exchange chemistries. MEPS is available in a variety of common SPE phases.
MEPS BARREL INSERT AND NEEDLE
The MEPS Barrel Insert and Needle (BIN) assembly contains the stationary phase, and is built into the syringe needle.
SAMPLE SIZE AND SENSITIVITY
TSample volumes may be as little as 10 μL, or by taking multiple aliquots of 100 μL or 250 μL, samples of 1 mL or larger may be concentrated.
Extract samples and make injections on-line using a single device, reducing sample processing times and the need for operator intervention.
EBIN life is dependent on the specific matrix being analyzed. For example, C18 analysis of whole plasma samples is conservatively 25-100 samples before the BIN needs to be changed. BIN life of cleaner samples is significantly longer.
Chromatogram of 22 pesticides standard solution (40 ng/mL)
The small quantity of phase in the MEPS BIN is easily and effectively washed between samples to reduce the possibility of carryover. This washing process is not practical with off-line SPE devices. With automation of MEPS, washing occurs while the previous sample is running.
FLEXIBLE AND EASY TO USE
The dimensions of the sorbent bed ensure performance remains identical to conventional SPE devices when used for extraction of similar samples. The AOC-MEPS system was developed to incorporate MEPS into a process automation workflow, combining sample preparation and analysis in a single platform
When automated by AOC-MEPS system, the injection volume is much larger compared to the commonly used GC injection volumes of 1 — 2 μL. With AOC-MEPS, the typical injections are 50 to 200 μL of elution solvent into the analytical instrument. A large volume injection volume technique should be employed that removes the solvent volume from inside the injector unit while condensing the target compounds.
PTV Inlet Liner for AOC-MEPS System
(AOC-20i + GC-2010 + PTV configuration)
|100 μL||✓||221-74830-01||Plunger Pk 1||221-74830-10|
|C8 + SCX||5||221-74830-05|
|Development Kit — contains 1 each of C18, C8, Silica, C8+SCX, C2||1 of each phase||221-74830-08|
|100 μL||✓||221-75202-01||Plunger Pk 1||221-75202-03|
|250 μL||✓||221-75202-02||Plunger Pk 1||221-75202-04|
|C8 + SCX||5||221-75198-03||221-75199-03|
|Development Kit — contains 1 each of
C18, C8, Silica, C8+SCX, C2
|1 of each phase||221-75198-05||221-75199-05|
For Research Use Only. Not for use in diagnostic procedures.
This page may contain references to products that are not available in your country.
Please contact us to check the availability of these products in your country.
Взято с http://web.archive.org/web/20200808064227/https://www.shimadzu.ru/spe-syringes